Descripción: manual electrocardiógrafo Cardiofax DECRETO LEGISLATIVO N° DEL REGIMEN DICIPLINARIO DE LA PNP (RESUMIDO). docx. D.L. DECRETO LEGISLATIVO QUE REGULA EL REGÍMEN DISCIPLINARIO DE LA PNP TITULO PRELIMINAR PRINCIPIOS. Grupo Ascenso PNP. LEY DEL RÉGIMEN DISCIPLINARIO DE LA PNP – DECRETO LEGISLATIVO Nº DEL 11 DE DICIEMBRE DE
|Published (Last):||15 March 2014|
|PDF File Size:||12.84 Mb|
|ePub File Size:||8.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
Any potential difference between instruments may cause electrical shock to the patient and operator. Please read these precautions thoroughly before attempting to operate the lehislativo. Format the SD memory card.
Manual Cardiofax 1150
Use only Nihon Kohden approved products dedreto this device. If the multiple portable socket outlet gets dusty, damaged or contacts liquid, it may cause instrument malfunction and electrical shock to the patient or operator.
The Resting ECG recording screen is displayed. Electrocardiograph overheated because it continuously recorded AC interference superimposed on the waveform.
decreto legislativo 1150 pdf
For details, refer to the MRI manual. The electrocardiograph could not Contact your Nihon Kohden representative. Date display format Select the date display format. CAL mark The height corresponds to 1 mV waveform amplitude. Potential difference between instruments may cause current to flow to the patient connected to the instruments, resulting in electrical shock micro shock. Only use new electrodes. There is a time delay between the input signal and output signal.
Also, noise may superimpose on the ECG waveforms and measurement data may be incorrect. The Data Management – Format screen appears. When the power is turned on, the Resting ECG recording screen appears. Enter the current date and time with the keyboard on the operation panel. Patient names on the screen shots and recording examples in this manual are fictional and any resemblance to any person living or dead is purely coincidental. Make sure that the electrode is completely dried after cleaning.
When the provided power cord cannot be used or when equipotential grounding is doubtful such as in a facility with poor groundingoperate the electrocardiograph on battery power.
decreto legislativo pdf – PDF Files
The lamp lights when copy is available and during printing. The LEDs on the operation panel light up one by one. Replace the battery with a legislatlvo one. Clean the skin with a cotton moistened with alcohol to remove oil. If the measured field strength in the location in which the ECG is used exceeds the applicable RF compliance level above, the ECG should be observed to verify normal operation.
Replace the built-in lithium battery. Enter the hospital name. The patient information is set and Resting ECG defreto screen appears.
Check that the recorder cover is firmly closed. Prints copies of the recording results.
Interference may occur in the vicinity of equipment marked with the following symbol: The System Settings screen is displayed. Connect the patient cable to the patient cable connector on the right side panel of the electrocardiograph. Off Battery charge lamp: The ECG data of the selected file is printed.
You can also perform automatic and manual recording during periodic ECG recording. A dirty thermal head degrades printing quality.
Avoid direct sprinkling, spray or decrero air from a nebulizer or humidifier. Pmp the MODE key. If abnormal performance is observed, additional measures may be necessary, such as re-orienting or relocating the ECG Before connecting or disconnecting instruments, make sure that each instrument is turned off and the power cord is disconnected from the AC socket.
At 80 MHz and MHz, the higher frequency range applies. In this case, temporarily changed settings are lost.
EQWEQ by Siete Tacsi Avila on Prezi
When it is off, the high cut filter is applied. Periodic recording starts with the preset settings. The changed setting is saved. The electrocardiograph does not print Excessive noise during printing causes the electrocardiograph does not anything or recording result is faint. Select the language for display and recording. Manage and operate the information appropriately.
Some of the panel keys also function as number keys on some screens. For details, refer to the ESU manual. The list includes measurement date, ID, gender, age and measurement mode.
Preparation for Manual Recording Change the following settings, if necessary. A large polarization voltage can have the following causes: