This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). Agilent Technologies. PSA Series Spectrum Analyzer. This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3.
|Published (Last):||11 June 2014|
|PDF File Size:||13.32 Mb|
|ePub File Size:||13.79 Mb|
|Price:||Free* [*Free Regsitration Required]|
Entry fields that are being modified or are modifiable are indicated by highlighting inverse video. Options see page Unless specified otherwise, all options are available when you order a spectrum analyzer; some options are also available as kits that you can order and install after you receive the analyzer.
Press If the previously performed alignments did not resolve the problem, press Alignments Restore Align Defaults Is the analyzer displaying an error message? Page 80 Index personality options not in instrument power amplifiers preamplifiers preset factory types user, creating printer setup probes AC and DC product information rear panel symbols rear panel features reference level, setting reference, turning on 10 MHz rename files RF limiters Place a marker labeled 1 on the 10 MHz peak, as shown in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and in the upper-right corner of the screen.
Power Requirements Power Requirements The only physical installation of your Agilent spectrum analyzer is a connection to a power source. Select the type of file you want to save.
In Case of Difficulty This chapter includes information on how to check for a problem with your Agilent Technologies spectrum analyzer, and how to return it for service. Select the type of file you want to rename setup, state, trace, limits, screens, or corrections. If so, refer to the Instrument Messages and Functional Tests manual.
Agilent Technologies E4440A Getting Started Manual
Got it, continue to print. Page 25 Item Description Data invalid indicator Status message area Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger or Trace Average Display line annotation In Maanual of Difficulty Returning an Analyzer for Service Returning an Analyzer for Agilemt Service Tag When you are returning an analyzer to Agilent Technologies for service, fill out and attach one of the blue service tags provided agilsnt the end of this chapter.
Reset the analyzer by pressing 2. File Menu Functions This chapter describes how to use the functions located under the front-panel Data storage and retrieval are handled similarly to that of personal computers PCs. F4440a Against Electrostatic Discharge Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen damage caused by ESD is present whenever components are transported, stored, or used.
Comments to this Manuals Your Name. Retain the shipping materials for future use, as you may wish to ship the analyzer to another location or to E44440a Technologies for service. Page 41 Improving Frequency Accuracy RS This feature not fully implemented; to be implemented in a future release. Restores the analyzer to its factory-defined state. Then press the desired terminator either a unit softkey, or the Terminate a value that requires s4440a unit-of-measurement.
Page 23 Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer. File Catalog Figure EA 3 Hz – This manual provides documentation for the following instruments:. A total of 3 years of warranty service support.
Agilent handheld spectrum analyzer user’s guide pages. WARNING If this product is to be energized via an external auto transformer for voltage reduction, make sure that its common terminal is connected to a neutral earthed pole of the power supply. This function can be auto-coupled. Wgilent analyzer does not contain customer serviceable fuses.
It is effective over a frequency range of dc to 2 GHz. Using The Front Panel Using the Front Panel Entering Data When setting measurement parameters, there are several ways to enter or modify the value of the active function: To access the Load menu, press 3.
The file cannot be placed inside another file. If your analyzer is covered by a separate maintenance agreement, please be familiar with its terms. Knob Increments or decrements the current value. To access the Copy menu, press boxes as described by the From: Preselection reduces mixer overload from broadband signals, reduces radiation of local e4404a harmonics back to the device under test, and reduces the level of image and multiple responses displayed.
Is the analyzer turned on?
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
To see that trace 2 is in view mode, blank traces 1 and 3 by pressing. When you have completed the printer setup for the analyzer, press the front panel key. Viewing a Signal 1.
To be implemented in future firmware upgrade. The front-panel keys are dark gray, light gray, green, or white in color.